Here's how the Aladin 2 Cassette works:
- In the Aisys Carestation, anaesthetic agent control consists of two parts: the electronic control mechanism in the anaesthesia system and the separate Aladin 2 agent cassette.
- Agent concentration is adjusted by regulating the amount of fresh gas flowing through the cassette. A proportional valve is used to regulate the flow. Part of the fresh gas bypasses the cassette, so the more fresh gas allowed to pass through the cassette, the greater the concentration.
- Fresh gas flow, along with flows in and out of the cassette, are electronically measured for enhanced control. Each Aladin 2 Cassette is magnetically coded, allowing the Aisys Carestation to recognize which type of agent cassette is inserted